I tried to get the VI above running, but getting a session handle error. Included measurement data is as following:. What does the block diagram look like? The control that says Instrument Handle on your front panel. The Softpanel measures and provides test report feature for recording all the key parameters such as Total Harmonics Distortion THD , waveform crest factor, voltage, current, frequency, active power, apparent power, power factor, etc.
|Date Added:||26 January 2014|
|File Size:||60.53 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The control that says Instrument Handle on your front panel.
Instead of traditional analog measurement circuits, the uses state-of-the-art DSP digitizing technology. Included measurement data is as following:. Chroma Power Meter is capable of meeting EN Yes, that is what is selectable in the list itself, together with com 3, com 4, com 5. Add to Inquiry Cart Data Chorma.
Can’t get Chroma working. – Discussion Forums – National Instruments
In addition to that, Chroma Power Meter is also capable of meeting the power measurement uncertainty requirements stated in section 4. The Softpanel measures and provides uxb report feature for recording all the key parameters such as Total Harmonics Distortion THDwaveform crest factor, voltage, current, frequency, active power, apparent power, power factor, etc.
Message 8 of The new standard has included interpretation, calculation and implementation methods for analyzing the measured power and the analyzing process is complex and time consuming.
Message 1 of Sub does the block diagram look like?
Digital Power Meter | Chroma ATE Inc.
I didn’t open the initialise VI before, I did try it a few times but don’t know how to use it: In order to resolve this, the Power Meter Softpanel has included the required data analysis feature and provide users with easy to use interface to carry out the standby power measurement according to the international standard.
Is actually what is selectable from the instrument handle pull-down? I have got the it working in my previous older computer in win7.
I installed the drivers: Message 2 of The instruments include a four part display with 7-segment LED front panel readouts. Wondering if anybody has a solution for this? I don’t know 662202 to look?
Message 7 of Can you describe more exact which handle you mean? Chroma Power Meter meets EN Message 10 of Included measurement data is as following: Did I make any mistake somewhere? Is the instrument handle you passed from the initialize VI or did you just open the fetch without doing the init? But those are virtual com’s to other hardware.
Instrument Control (GPIB, Serial, VISA, IVI)
Message 3 of I just noticed you have just the fetch VI open. Can’t get Chroma working.
Users can easily select desired parameters 6622 readouts at a touch of a button. I tried to get the VI above running, but getting a session handle error.
Message 6 of The Model includes simple measurement functions designed for low power applications maximum current 2A.