Since the repair I have been using the power supply for a little over a week and I have had no issues whatsoever, and I am very happy how it turned out. Leave a Reply Cancel reply. Page 33 Front panel binding posts After it is loose the cover tilts forward and is easily removed. Refer to the Service Information for more information. Request a Calibration Quote.
|Date Added:||13 November 2018|
|File Size:||69.73 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent EA Power Supply – File Exchange – MATLAB Central
Page Self-Test Procedures Page 36 Output Connections By making it simple to connect, control instruments, and automate test sequences you can quickly move past the test development phase and access results faster than ever before with just a few clicks. Because the encoder snaps into place I had to cut the two large legs off of the old encoder before I agiilent comfortably remove it from the board.
Rent From Transcat Do you need a test instrument for a short time? Page 27 To Rack Mount the Instrument Finally the board is out and the rotary encoder can be replaced!
Repair of an Agilent EA DC Power Supply – William Breidenthal
Page Execution Errors zero Skip to content I recently bought a used Agilent power supply in a very sad condition. Transcat makes it easy: Would you like this item calibrated?
New Purchase Calibration Learn More Save time and money by calibrating your new instrument before it leaves our warehouse, ensuring it arrives ready to be put into service. It does have two speeds loud-ish and louder?
Page 90 Calibration Commands Bit Definitions — D3640a Event Register Page Operating Checklist Operating Checklist The underside of the board has some giant copper traces, and a large portion has the silkscreen removed and has been tinned.
Triggering Notice that the time delay is valid for only the bus trigger source. To print the manual completely, please, download it.
I recently bought a used Agilent power supply in a very sad condition. This is actually towards the front of the power supply making it a little awkward to pick up. Page End of Program Ships in weeks. Page 38 Output Connections Unreg After doing a little research I found that it was a known issue for the rotary encoders in these power supplies to fail.
Page Troubleshooting Hints After removing a couple of screws the whole inside of the e364a0 supply slides out the front. Page 8 In This Book A couple days later I had a new rotary encoder ready to be installed.
W3640a Continued on next page Finally on the other side where the PCB is the tab routed in the board needs to be lifted gently with a small screwdriver until it clears the white plastic tab.